Connecting to Bus Topologies - CableIQ
Connecting to Bus Topologies
Telephone cables wired in a bus topology (below) connect the wall outlets in series. In this topology, you measure the length from the last outlet to the distribution center.
If you connect to an outlet in the middle of the series, the tester reports a bridge tap. The length reported is the length to the outlet, which is the patch cord length. The tester cannot measure length past the outlet because reflections from the cables on either side interfere with measurements.
Tip: To quickly verify the wire maps of telephone cabling connected to a bridge tap, use the MultiMap function in DISCOVER mode.
If you are unsure which outlet is the last in the bus, do the following: